Paul FrameThank you very much for all your help regarding the SADT Concrete Hammer and hardness tester which are working well and we are very pleased with it.
Iso Digital Shore Hardness Tester Convenient Holding Measured Display Values
|Flat Cone Point (0.79 Mm) 35º Included Angle
|Length Of Probe
iso digital rubber hardness tester,
iso digital shore hardness tester,
8.1N digital shore hardness tester
Digital Rubber Hardness Tester Shore A/C/D
Shore digital rubber hardness tester is designed for testing the hardness of soft rubber, printer roller and other elastomer material.(ASTM--D--2240). It has advantages of simple structure, convenient use, small body and light weight, intuitive readings. It can carry the hand-held measurements and can also be installed on the same type of fixed cradle which is matched with the set load test.
Very convenient to use
Easy to carry
Holding measured display values
Power Supply: one 1.5 V cell battery
Used to test sof t rubber, prnter roller and other elastomer materials
Graduation: 0-100 degree
Indentor press on telos: A, C type 0.55N-8.06N, D type0-44.5N
|general rubber, synthetic rubber, soft rubber, poly-resin, leather, wax, etc.
|rubber and contains the tiny hole material that that the vesicant is made in being used for plastic
|general hard rubber, hard resin, acryl, glass, thermoplastics, printing plates, fibres, etc.
|Flat cone point 0.79mm 35º Included Angle
|Sharp cone point SR 0.1mm 30º Included Angle
Make zero before measure it . when durometer press on the glass ,the hand should be on the position of 100 degree. Press the hand several time if it is not on zero or 100 degree.I f ure on the rack, can lift the hand shank,make the durometer under the gravity of the poise , the press part and the table contact completely , the reading should be 100+1(degree), if it is not 100+1(degree),use the hand wheel to adjust it . If it is still not workable. pls send to the manufacturer.